# Digital atom interferometer with single particle control on a discretized space-time geometry

See allHide authors and affiliations

Edited by Peter Zoller, University of Innsbruck, Austria, and approved April 19, 2012 (received for review March 14, 2012)

## Abstract

Engineering quantum particle systems, such as quantum simulators and quantum cellular automata, relies on full coherent control of quantum paths at the single particle level. Here we present an atom interferometer operating with single trapped atoms, where single particle wave packets are controlled through spin-dependent potentials. The interferometer is constructed from a sequence of discrete operations based on a set of elementary building blocks, which permit composing arbitrary interferometer geometries in a digital manner. We use this modularity to devise a space-time analogue of the well-known spin echo technique, yielding insight into decoherence mechanisms. We also demonstrate mesoscopic delocalization of single atoms with a separation-to-localization ratio exceeding 500; this result suggests their utilization beyond quantum logic applications as nano-resolution quantum probes in precision measurements, being able to measure potential gradients with precision 5 × 10^{-4} in units of gravitational acceleration *g*.

## Localized Particles

Particle interferometry has yielded numerous important results from the advent of quantum mechanics until now; counting only the most recent ones, they range from precision measurements (1) and fundamental tests of quantum mechanics (2, 3) to applications in optical and superconducting magnetometry (4, 5). Whereas other architectures provide a natural way to engineer, guide, and confine interfering paths, e.g., by means of optical waveguides for photons or superconducting circuits for electrons, one key challenge for atoms consists in realizing a flexible platform that is capable of providing complex interfering geometries where the atoms remain localized at all times. Unprecedented control over atomic states has been obtained in optical lattice potentials; designed to mimic solid state systems they resemble engineered circuits. To construct interfering paths in these systems, an additional quantum number is needed—that has to be decoupled from motional states—for the atomic wave packet’s motion to be steered in a coherent and state-dependent manner. This additional control is achieved by leaving the simple scalar concept of an atom and taking advantage of its vector nature endowed by electronic and nuclear spin. Mediated by the spin-orbit interaction, optical dipole forces yield a state-dependent force, offering the right handle for steering atomic paths. Therefore, an effective magnetic field coupled to the atomic magnetic moment makes position control of the interfering paths possible, albeit at the expense of an increased sensitivity to magnetic fields (6⇓–8). Even though this direct coupling to magnetic fields is not a fundamental limitation, it represents a technical hurdle that demands efficient shielding against stray magnetic field fluctuations; accepting this price, digital engineering of atomic paths in optical potentials can bring atom interferometry to a unique level of control.

Particle localization and digital reprogrammability of atomic paths, besides being necessary resources in quantum technologies, e.g., for quantum cellular automata (9) and multiparticle entanglement (10), specify also the application range where trapped-atom interferometers could outperform other interferometers in precision measurements. In general, to measure constant homogeneous potential gradients the figure of merit is the space-time area enclosed by the atomic paths, and the ultimate precision is fixed by shot noise as standard quantum limit. Although space-time areas achieved by free-fall atom interferometers, e.g., of about 1 mm × 160 ms in ref. 1, are in principle reachable by our interferometer, and at the moment only technically limited, the prime advantage in having trapped atoms resides in the nanoscale spatial resolution, allowing nondestructive probing of microscopic atomic quantum systems and measuring potential gradients at ultrashort scale.

## Interferometer

Atom interferometers operate by splitting the wavefunction of an atom onto two spatially separated paths, and measuring their phase difference (11, 12). As depicted in Fig. 1*A*, for this purpose, the atom is prepared in a quantum superposition of states; these are subsequently delocalized and eventually coherently recombined to probe the phase difference accumulated between the two paths. Our interferometer uses two long-lived internal states of cold Caesium atoms, and . Although these internal states are coherently manipulated by means of microwave pulses, the particle position is controlled by trapping atoms in a one-dimensional optical lattice with spacing *d* = 433 nm. The spin-dependent control is realized through a lin-*θ*-lin configuration of the optical lattice, which allows atoms to move in discrete steps in a direction dependent on the spin state , by ramping the polarization angle *θ* (see *Materials and Methods*) (13, 14). When a superposition of the two spin states is created by a *π*/2 pulse, the trapped atom is delocalized by a single step in a fully coherent manner. As illustrated in Fig. 1*B*, shift operations of alternated direction are interleaved with *π* pulses to further separate the paths. Up to 100 block operations can be concatenated in one coherent digital chain, eventually achieving coherent separations larger than 10 μm, as shown in Fig. 1*C*. The tight confinement by the lattice potential ensures that each wavefunction’s component is axially strongly localized in space, down to 18 nm, achieving at the same time nanoscale positioning control and mesoscopic spatial separation.

To extract the interferometric phase difference Φ between the two paths, we employ a Ramsey probing scheme where the final merging block maps the phase information onto the two spin populations by means of a *π*/2 pulse of variable phase *φ*. Scanning the phase varies the probability to detect [1]γ being the probability of an atom escaping from the trap during the interferometer sequence, and *C* the Ramsey fringe contrast. When the interferometer is operated with a single atom, detecting the presence of the atom in, e.g., state results in a 0/1 binary signal at each run of the sequence, as displayed in Fig. 1*D*. Unlike with large ensembles of atoms, to measure the underlying probability distribution in Fig. 1*E* multiple repetitions are here averaged in time.

## Geometries

The geometry of the interferometer is determined by the programmed sequence of operation blocks. The elementary diamond geometry, which resembles the classic Mach–Zehnder interferometer, maximizes the sensitivity of the interferometric phase to potential gradients. Varying the size of the diamond reveals how phase and coherence of the superposition state evolve with the number of shifts. Fig. 2*A* shows a strong parabolic phase accumulation, which we can track over 11*π* radians. Such a phase behavior is expected in the presence of a linear potential gradient ∇*U*, as it can be interpreted as the gradient strength times the space-time area that the paths enclose. The accumulated phase will then follow [2]*n* being the total number of shift operations, *τ*_{S} the duration of a shift block, and *τ*_{π} the duration of a *π* pulse, and *x*_{L}, *x*_{R} the coordinate along the left and right path, respectively. The phase evolution in Eq. **2** has been fitted to Fig. 2*A*, indicating a potential gradient of 2*πℏ* × (324.5 ± 0.8) Hz/*d*, equivalent in magnitude to a gravitational acceleration of (0.2296 ± 5·10^{-4}) *g*. We attribute this gradient to a displacement of the lattice laser focus from the position of the atoms by ≈600 μm, about a quarter of the Rayleigh length; the beam’s divergence leads to a reduction of the optical trap depth further away from the focus, thus creating a potential gradient which is nearly linear over the experimental region of ≈40 μm. This hypothesis is supported by the direct proportionality measured between the gradient strength and the trapping laser’s power, as shown in Fig. 2*A*, *Inset*.

To cancel the phase caused by such a background gradient, we consider a double diamond geometry, consisting of two basic diamond interferometers interspaced with a *π* pulse, see Fig. 2*B*. By crossing the paths in the middle of the sequence, the second loop accumulates a phase opposite to the first loop, and cancellation occurs akin to a spin echo. This geometric spin echo is effective for any spin-and time-independent potential. Gradients which are active only for one half of the sequence are still detected as described above, making the double diamond interferometer suitable to measure time-controllable gradients in a differential scheme. The successful suppression of the parabolic phase is visible in Fig. 2*A*.

## Decoherence

To mitigate decoherence effects the interferometer block chain is assembled such that each subsequence of shift, *π* pulse, shift operations achieves on its own a spin echo refocusing, compensating spin-dependent inhomogeneous disturbances; such repeated spin echoes are reminiscent of dynamical decoupling (15, 16). Fig. 2*C* shows an exponential decay of contrast for increasing interferometer size, which we observed for both interferometer geometries. This behavior indicates that identically repeated operation blocks are subject to an invariable decoherence per block, suggesting the number of shifts as the relevant quantity for contrast.

To isolate the cause of this decay, we repeat the same sequence replacing all the shift operations by idle blocks, and fit the contrast decay with an exponential curve (dashed line) showing a contrast loss of 0.6% per step. Decoherence in this case is dominated by time-varying differential light shift due to fluctuations of the lattice laser power; in addition, because spin-dependent transport excludes the use of clock states, a minor contribution from magnetic field fluctuations is also expected.

The significantly lower decay rate without delocalization on the lattice (dark green area) indicates that most decoherence in the interferometer takes place because of shift operations (light green area). We hence measure independently the transport efficiency per single shift step ± *d*/2 using methods of ref. 17, and, after preparing atoms in the axial motional ground state (18), we achieve 99% success probability for shifting the atom in the correct direction, limited by the efficiency of the microwave pulses. Accounting for this probability once per shift and per path, the shift-free decay rate is combined with a further 2% contrast drop per step to produce the dash-dotted curve in Fig. 2*C*. We attribute the residual 1.7% contrast drop per step to polarization jitter during shift operations and, to a smaller extent, to spurious motional excitations.

## Hold Times

One distinguished feature of a trapped particle interferometer is the ability to insert an arbitrary hold time, *t*_{hold}, in between blocks, where the interferometer is held open at a fixed separation. This capability becomes relevant when the interferometer is to be operated near an object of study, with long interaction time and nanoscale control of distance, e.g., a surface or even a Bose–Einstein condensate. The natural place to insert a hold time is at maximum separation of the paths, see Fig. 3*A*; this situation is investigated in Fig. 3*B* for sequences of *n* = 4, 8, and 12 shifts where the interferometric phase is measured varying *t*_{hold}. In the presence of a potential gradient an additional phase Φ_{hold} = ∇*U*·*ndt*_{hold}/(2*ℏ*) arises from the space-time area added by the hold time, on top of the phase in Eq. **2**. A linear fit yields a potential gradient of 2*πℏ* × (324 ± 7) Hz/*d*, in full agreement with the results in Fig. 2.

To protect the superposition state from dephasing during the hold time two *π* pulses are applied, realizing a twofold spin echo refocusing, see Fig. 3*A*. The fact that the phase Φ_{hold} survives spin echoes confirms the spin-independent nature of the potential gradient, as expected from such a light shift gradient. As shown in Fig. 3*C*, the contrast as a function of the hold time reveals a Gaussian decay (19) on a timescale comparable to that of nontransported atoms.

## Inertial Force Measurement

An important application of atom interferometry is the measurement of external forces. To demonstrate this capability, we apply a controlled inertial force by accelerating the whole lattice potential during a fixed acceleration time *t*_{acc} (see *Materials and Methods*). As shown in Fig. 4*A*, the paths are then recombined in the moving inertial frame, and the effect of the force *ma* is measured from the interferometric phase, with *m* the atomic mass and *a* the applied acceleration. We employ the simplest geometry of the single diamond and find that the phase linearly increases with the applied acceleration *a* with a slope proportional to path separation (Fig. 4*B*). Calculating the space-time area in this case yields a phase of Φ_{acc} = *ma*·*ndt*_{acc}/(2*ℏ*) for *n* total shifts, where *a* is independently calibrated. The agreement up to 5 *g*—only technically limited—together with the achieved precision of 5 × 10^{-4} *g* in Fig. 2 demonstrate the ability of the single-atom interferometer to measure external fields in real applications, showing a dynamic range of at least four orders of magnitude.

## Conclusions and Outlook

In summary, we have demonstrated an atom interferometer operating at the single-atom level; we proved its capability to measure external fields—both the divergence of an optical dipole trap and an externally controlled inertial force—achieving a resolution which borders precision applications. By exploiting the block modularity we could precisely identify the decoherence mechanisms and explain the loss of contrast in terms of decoherence per shift operation. Yet, the full digital control established here allowed us to suppress decoherence effects by engineering the interferometer’s paths to effect a geometrical refocussing.

In the quest for quantum devices functioning at the single-atom level, from Rydberg quantum logic gates (20, 21) to single-site-resolved optical lattices (22⇓–24), our results lay the base for digital quantum simulators (25) and quantum cellular automata (9, 26) based on interacting atoms in lattice potentials (27, 28). Similarly, scalable arrays of quantum logic gates with single-atom addressing are also within reach (10). Further, this system promises to reveal quantum coalescence of massive boson particles in a two-atom Hong–Ou–Mandel interference (29), with noticeable implications for sub-shot-noise interferometry (30, 31).

The nanoscale single-atom control evidenced here is particularly suitable for the investigation of surface effects, such as Casimir–Polder interactions or Yukawa-like gravitational potentials at micrometer distances (32). Reducing technical sources of dephasing, we expect 10-fold improvement of coherence time, allowing measurement of Casimir–Polder potential with single atoms with 5% precision at distances shorter than 4 μm from a dielectric surface, assuming 24 h integration time (33). Furthermore, an implementation of conveyor belts for spin-dependent optical lattices is currently underway, with which we expect to reach coherent splitting distances of the order of 1 mm (34). Combining this large separation with a hold-time scheme for interrogation in decoherence-protected clock states, we deem realistic an improvement of space-time area by a factor 10^{5} over the present situation, eventually rivaling in precision free-fall atom interferometers.

## Materials and Methods

### Lattice.

Two counterpropagating linearly polarized laser beams at wavelength *λ* = 866 nm, with the polarization of the retroreflected beam rotated by a variable angle *θ* (lin-*θ*-lin configuration), produce an optical lattice potential with spacing *d* = *λ*/2; the beams are focused to the position of the atoms with a Rayleigh length of 2.3 mm. The trap frequencies of each potential well are *ω*_{ax} = 2*π* × 120 kHz,*ω*_{rad} = 2*π* × 2 kHz and the off-resonant scattering rate is 10 Hz. After molasses cooling to *T* = 10 μK, atoms are cooled in the lattice to the motional ground state along the axis (18). The two spin states, and , experience different light shifts induced by the left and right circular components of the laser, which allows spin-dependent shift operations by controlling the angle *θ* with an electro-optic modulator (7, 8, 13). The beams’ polarization is controlled with a relative precision of 10^{-4}, and the shift duration is set depending on *ω*_{ax} to suppress motional excitations (13).

### Detection.

The interferometric phase is mapped to spin populations by a *π*/2 pulse around the variable axis . The spin population is detected by selectively pushing out the atom when in the state , and subsequently detecting the presence of an atom by fluorescence image; atom losses *γ* ≈ 5% due to background collisions and light scattering during fluorescence imaging cannot be distinguished from state .

### Coherence Time.

The bare spin coherence time is *T*_{2} ≈ 200 μs, which is mostly determined by radial thermal motion causing inhomogeneous variations of the differential light shift of the two hyperfine states (19). One spin echo pulse is able to suppress such an inhomogeneous dephasing, resulting in an extended coherence time . Spin echo series as in the interferometer sequence benefit from a dynamical decoupling effect (15, 16), yielding a longer up to 2.3 ms. This time should be compared to the single step duration of *τ*_{S} = 18 μs per shift plus *τ*_{π} = 12 μs per *π* pulse.

### Scalability.

To increase the acquisition rate several interferometers are executed in parallel by loading on average about 30 atoms into the dipole trap over a region of 40 μm (Figs. 2⇑–4), where each run of the interferometer lasts about 1.5 s. Because of the weak radial confinement, atom-atom interactions are negligible.

### Inertial Force.

An inertial force is created by accelerating the lattice potential; we apply a parabolic voltage ramp to a piezo actuator holding the retroreflecting mirror. The displacement-to-voltage dependence is calibrated with a Michelson interferometer, producing the mean acceleration ∫*a*(*t*)d*t*/*t*_{acc} used in phase calculations. Methods from ref. 35 raise the first piezo resonance to over 50 kHz; maximal acceleration is bounded by electronic limitations. Interband Landau–Zener tunneling is negligible below the critical acceleration of 50 × 10^{3} m/s^{2}.

## Acknowledgments

We thank Reinhard Werner for discussions. We acknowledge support from the Deutsche Forschungsgemeinschaft FOR635, the European Commission’s Scalable Quantum Computing with Light and Atoms (SCALA) and Atomic Quantum Technologies (AQUTE) projects. A. Steffen acknowledges support from the Studienstiftung des deutschen Volkes. A. Alberti acknowledges support by the A. v. Humboldt Foundation. A. Steffen and S. Hild acknowledge support from Bonn–Cologne graduate school.

## Footnotes

- ↵
^{1}To whom correspondence should be addressed. E-mail: alberti{at}iap.uni-bonn.de.

Author contributions: W.A., A.W., and D.M. designed research; A.S., A.A., N.B., S.H., and M.K. performed research; A.S. analyzed data; and A.S. and A.A. wrote the paper.

The authors declare no conflict of interest.

This article is a PNAS Direct Submission.

## References

- ↵
- ↵
- ↵
- ↵
- ↵
- Clarke J,
- Braginski AI

- ↵
- ↵
- ↵
- ↵
- Grössing G,
- Zeilinger A

- ↵
- ↵
- Berman PR

- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- ↵
- Lloyd S

- ↵
- Lloyd S

- ↵
- Karski M,
- et al.

- ↵
- Ahlbrecht A,
- et al.

- ↵
- ↵
- Giovannetti V,
- Lloyd S,
- Maccone L

- ↵
- Dunningham J,
- Rau A,
- Burnett K

- ↵
- ↵
- ↵
- Kuhr S,
- et al.

- ↵

## Citation Manager Formats

## Article Classifications

- Physical Sciences
- Physics